|Tank Size:||500 X 300 X 200 Mm||Unit Size:||530 X 325 X 325 Mm|
|Paking Size:||635 X 405 X 390mm||Transduces:||10pcs|
|Ultrasonic Power:||600W||Timer:||0-30 Minutes Adjustable|
|G.W:||16KG||Power Supply:||AC 110V ; AC 220V|
30min Timer Mechanical Ultrasonic Cleaner,
30L Mechanical Ultrasonic Cleaner,
SUS304 Portable Ultrasonic Cleaner
Mechanical Knob Ultrasonic Cleaner Ultrasonic Surgical Instrument Cleaning Bath
|Material||Stainless Steel SUS304|
|Timer||1-30minutes , Mechanical timer|
|Heater||20°C to 80°C, Mechanical Heating|
|Power Supply||AC 100 ~ 120V, 50 / 60Hz|
|AC 220 ~ 240V, 50 / 60 Hz|
|Ultrasonic Power||600 W|
|Heating Power||500 W|
|Tank Size||500 x 300 x 200 mm ( L x W x H )|
|Unit Size||530 x 325 x 325 mm ( L x W x H )|
|Package Size||635 x 405 x 390 mm ( L x W x H )|
Suggested Ultrasonic Cleaning Procedure
In all cases manufacturers’ instructions should be followed when using an ultrasonic cleaning process. These are representative steps.
Fill the ultrasonic cleaning tank with an approved medical instrument cleaning solution such as CLN-LR012 available from Tovatech following dilution instructions provided. Turn the cleaner on to start the degassing process. This step removes entrained air in new solutions that interferes with the efficiency of cavitation and takes approximately 10 minutes.
In the meantime:
At the end of the cycle, remove the instruments from the ultrasonic cleaning bath and thoroughly rinse them to remove all traces of the cleaning solution. Deionized water rinses will avoid spotting. If the instruments are not to be immediately disinfected and sterilized be certain that they are thoroughly dried and protected. Part reassembly can occur after sterilization.
Procedures should be in place to guide the replacement of used ultrasonic cleaning solutions. In some instances it is recommended that solution be drained and tanks thoroughly cleaned and dried after each ultrasonic cleaning cycle. Most solutions available today are biodegradable, which facilitates disposal but local authorities should be consulted on proper practices.
The ultrasonic cleaner uses ultrasonic waves (vibration) using water with detergents or enzymatic products to break up soil and organic material on medical instruments/devices. These devices are rinsed then autoclaved (sterilized). The autoclave sterilizer uses heat, steam, and pressure to kill all pathogenic microorganisms and their spores.
1 x Digital Ultrasonic Cleaner
1 x Cleaning Basket
1 x Manual
Contact Person: Crystal