Payment & Shipping Terms:
|Model:||JTS-3024GPH||First Tank Volume:||38L|
|First Tank Size:||500X300X250mm||Transducers:||12pcs In 1st Tank; 12pcs In 2nd Tank|
|Ultrasonic Power:||600W*2||Heating Power:||1500W*2|
3 Bins Ultrasonic Cleaner for Surgical knife blade/tools with Rinsing, Filter, Dryer 38L
|1st Tank Volume||38L Ultrasonic Cleaning tank with filtration cycle system|
|2nd Tank Volume||38L Ultrasonic Rinsng tank|
|3rd Tank Volume||38L Drying tank|
|1st tank size||500X300X250mm|
|2nd tank size||550X400X250mm|
|3rd Tank size||550X400X250mm|
|Transducers||12pcs in 1st tank; 12pcs in 2nd tank|
|Heating Power of Ultrasonic tank||1500W*2|
|Heating Power of Drying tank||750W|
|Drying System||1 set|
|Timer||0~99 minutes adjustable|
|Heater||Room temperature~99℃ adjustable|
|Material||SUS 304 default, can be SUS 316|
|Power Supply||AC 220V 3 phase ; AC 380V 3 phase|
280KG Wrapped by film in wooden case
|Lead time||18 working days|
|Warranty||1 year warranty period, technique support all time|
|Certificates||CE, RoHS, FCC, SGS|
|Advantage||Filter the dirty water; keep the water clear in tank; lower the frequency of changing water|
Suggested Ultrasonic Cleaning Procedure
In all cases manufacturers’ instructions should be followed when using an ultrasonic cleaning process. These are representative steps.
Fill the ultrasonic cleaning tank with an approved medical instrument cleaning solution such as CLN-LR012 available from Tovatech following dilution instructions provided. Turn the cleaner on to start the degassing process. This step removes entrained air in new solutions that interferes with the efficiency of cavitation and takes approximately 10 minutes.
In the meantime:
Segregate instruments by alloy or composition to avoid potential damage (Chromium plated instruments should not be cleaned ultrasonically)
Instruments with movable parts should be disassembled to facilitate cleaning
Place the instruments the ultrasonic cleaner’s mesh basket, taking care that they do not come in contact with each other
Cannulated or lumened instruments should be positioned to insure interiors are wetted with the cleaning solution. In some instances placing them on an angle will facilitate this
Set the control panel per manufacturers’ instructions and start the cleaning process
At the end of the cycle, remove the instruments from the ultrasonic cleaning bath and thoroughly rinse them to remove all traces of the cleaning solution. Deionized water rinses will avoid spotting. If the instruments are not to be immediately disinfected and sterilized be certain that they are thoroughly dried and protected. Part reassembly can occur after sterilization.
Procedures should be in place to guide the replacement of used ultrasonic cleaning solutions. In some instances it is recommended that solution be drained and tanks thoroughly cleaned and dried after each ultrasonic cleaning cycle. Most solutions available today are biodegradable, which facilitates disposal but local authorities should be consulted on proper practices.
Same series with different size for your choice:
|Model||Size of every tank||Tank Volume||Ultrasonic
|Heating Power of Ultrasonic tank||Heating Power of drying tank||Transducer|
How do I know what size of ultrasonic cleaner is best for my operation?
The two points you should think about,
● How about your cleaning volume? For very little small part, a table top ultrasonic are able to do this job. For large size and huge volume, you should consider industrial ultrasonic cleaner , for highest cleanness level , multi-stage ultrasonic cleaning system are recommend.
● The size of cleaning object, it can fit in the basket fully immersed under water.
Skymen Cleaning Equipment Shenzhen Co., Ltd., was founded in 2007,which follows the brand development strategy of “based in Shenzhen, radiating throughout the country, expanding overseas”.
Skymen also owns a professional R&D team to keep the pace with international ultrasonic technology. A passionate and young sales team sell to the world through different channels.
Contact Person: Crystal